Helmholtz Coils
HelmholtzCoils are especially designed to generate precisely defined magnetic fields from DC to the upper end of the audio frequency range and beyond. The generated fields are in a strongly linear relation to the coil current. The field strength can be calculated exactly by analytical (or numerical) methods, based on the coils' geometry, the number of turns and the coil current.
Therefore HelmholtzCoils are ideally suited for the calibration of magnetic field probes or sensors. Typical applications are magnetic immunity testing according to automotive standards or MIL461F.
When generating magnetic fields with Helmholtz coils the coil current is directly proportional to the magnetic field strength.
 Details
HHS 52016  Circular Helmholtz Coils


Number of turns: 
6 

Maximum Coil Current: 
40 A (1 min.) 15 A (15 min.) 

Nominal Coil Current: 
12 A continuous 

Coil Spacings (centered): 
50 ... 80 mm 




The following data refers to a coil separation of 60 mm 




Max. Magn. Field Strength: 
2860 A/m (1 min.) 

Nominal Magn. Field Strength: 
860 A/m continuous 

Magn. Field Strength, 1 A Coil Current: 
71.55 A/m


Coil Diameter: 
120 mm 

Mechanical Dimensions: 
290 x 220 x 130 mm 

Terminals: 
4 mm female 

Inductance (Single Coil): 
8 µH 

Inductance (Coil Pair): 
18.7 µH / 50 mm 18.2 µH / 60 mm 17.9 µH / 70 mm


Usable Frequency Range: 
DC  5 MHz 

Resonance Frequency: 
> 8 MHz 

Weight:

2.8 kg

Datasheet HHS 52016


 Details
HHS 520198  Circular Helmholtz Coils


Number of turns: 
98 

Maximum Coil Current: 
40 A (1 min.) 15 A (15 min.) 

Nominal Coil Current: 
12 A continuous 

Coil Spacings (centered): 
approx. 29 ... 95 mm 

Max. Magn. Field Strength: 
64 kA/m (1 min.) 

Nominal Magn. Field Strength: 
20 kA/m continuous 

Magn. Field Strength, 1 A Coil Current: 
1598 A/m


Conversion CurrentFieldstrength: 
64.07 dB/m


Coil Diameter: 
84 mm (58110 mm) 

Free Coil Distance for DuT: 
0 ... 66 mm 

Mechanical Dimensions: 
240 x 240 x 130 mm 

Terminals: 
4 mm female 

Inductance (Single Coil): 
0.84 mH 

Inductance (Coil Pair): 
2.3 mH / 30 mm 2.0 mH / 50 mm 1.8 mH / 90 mm


Usable Frequency Range: 
DC  200 kHz 

Resonance Frequency: 
> 600 kHz 

Weight:

4.1 kg

Datasheet HHS 520198


 Details
HHS 52029  Circular Helmholtz Coils


Number of turns: 
9 

Maximum Coil Current: 
55 A (5 min.) 

Nominal Coil Current: 
33 A continuous 

Coil Spacings (centered): 
116 mm / 132 mm 




The following data refers to a coil separation of 116 mm 




Max. Magn. Field Strength: 
3053 A/m (5 min.) 

Nominal Magn. Field Strength: 
1832 A/m continuous 

Magn. Field Strength, 1 A Coil Current: 
55.52 A/m 154.89 dBµA/m


Coil Diameter: 
232 mm 

Mechanical Dimensions: 
0.20 m x 0.25 m x 0.39 m 

Max. cubical shaped DuT: 
11.3 cm x 11.3 cm x 11.3 cm 

Terminals: 
4 mm female jacks 7 mm screw terminals 

Inductance (Single Coil): 
36 µH 

Inductance (Coil Pair): 
80 µH


Usable Frequency Range: 
DC  2.5 MHz 

Resonance Frequency: 
> 3.5 MHz 

Weight:

4.9 kg

Datasheet HHS 52029


 Details
HHS 520281  Circular Helmholtz Coils


Number of turns: 
81 

Maximum Coil Current: 
6 A (5 min.) 

Nominal Coil Current: 
5 A continuous 

Coil Spacings (centered): 
116 mm / 132 mm 




The following data refers to a coil separation of 116 mm 




Max. Magn. Field Strength: 
3000 A/m (5 min.) 

Nominal Magn. Field Strength: 
2000 A/m continuous 

Magn. Field Strength, 1 A Coil Current: 
500A/m 174 dBµA/m


Coil Diameter: 
232 mm 

Mechanical Dimensions: 
0.20 m x 0.25 m x 0.39 m 

Max. cubical shaped DuT: 
11.3 cm x 11.3 cm x 11.3 cm 

Terminals: 
4 mm female 7 mm screw terminals 

Inductance (Single Coil): 
2.83 mH 

Inductance (Coil Pair): 
6.42 mH


Usable Frequency Range: 
DC  300 kHz 

Resonance Frequency: 
> 400 kHz 

Weight:

4.7 kg

Datasheet HHS 520281


 Details
HHS 520412  Circular Helmholtz Coils


Number of turns: 
12 

Maximum Coil Current: 
60 A (5 min.) 

Nominal Coil Current: 
30 A continuous 

Coil Spacings: 
200 mm, 250 mm, 300 mm 



The following data corresponds to a coil spacing of 200 mm 



Max. Magn. Field Strength: 
2500 A/m (5 min.) 

Nominal Magn. Field Strength: 
1288 A/m continuous 

Magn. Field Strength, 1 A Coil Current: 
42.93 A/m 152.66 dBµA/m


Conversion Current>Fieldstrength:

32.66 dB/m 

Coil Diameter: 
400 mm 

Mechanical Dimensions: 
0.38 m x 0.58 m x 0.42 m 

Terminals: 
4 mm female 

Inductance (Single Coil): 
120 µH 

Inductance (Coil Pair): 
2.40  2.60 µH 

Usable Frequency Range: 
DC  500 kHz 

Resonance Frequency: 
>1.4 MHz 

Weight:

10.2 kg

Datasheet HHS 520412
files for the MagTest software


 Details
HHS 520436  Circular Helmholtz Coils


Number of turns: 
36 

Maximum Coil Current: 
20 A (5 min.) 

Nominal Coil Current: 
10 A continuous 

Coil Spacings: 
200 mm, 250 mm, 300 mm 



The following data corresponds to a coil spacing of 200 mm 



Max. Magn. Field Strength: 
2500 A/m (5 min.) 

Nominal Magn. Field Strength: 
1288 A/m continuous 

Magn. Field Strength, 1 A Coil Current: 
128.8 A/m 162.2 dBµA/m


Conversion Current>Fieldstrength:

42.2 dB/m 

Coil Diameter: 
400 mm 

Mechanical Dimensions: 
0.38 m x 0.58 m x 0.42 m 

Terminals: 
4 mm female 

Inductance (Single Coil): 
1.15 mH 

Inductance (Coil Pair): 
2.56 mH 

Usable Frequency Range: 
DC  150 kHz 

Resonance Frequency: 
>200 kHz 

Weight:

8.5 kg

Datasheet HHS 520436


 Details
HHS 52068  Circular Helmholtz Coils


Number of turns: 
8 

Maximum Coil Current: 
55 A (5 min.) 

Nominal Coil Current: 
34 A continuous 

Coil Spacings (centered): 
300 mm 

Max. Magn. Field Strength: 
1060 A/m (5 min.) 

Nominal Magn. Field Strength: 
650 A/m continuous 

Magn. Field Strength, 1 A Coil Current: 
19.23 A/m 145.68 dBµA/m


Coil Diameter: 
600 mm 

Max. Cubical shaped DuT: 
32.5 x 32.5 x 32.5 cm 

Mechanical Dimensions: 
0.64 m x 0.79 m x 0.42 m 

Terminals: 
4 mm female 

Inductance (Single Coil): 
91 µH 

Inductance (Coil Pair): 
200 µH 

Usable Frequency Range: 
DC  800 kHz 

Resonance Frequency: 
> 1500 kHz 

Weight:

17.0 kg


Recommended accessories:

NFCN 9732 85 kHz

Datasheet HHS 52068
Recommended accessories:


NFCN 9732 85 kHz  Compensation Network


The compensation network NFCN 9732 85 kHz consists of a high current/ high voltage variable capacitor.
The compensation capacitor reduces the total impedance of the series circuitry consisting of the Helmholtz coil HHS 52068 and the NFCN 9732 85 kHz at the operating frequency band of 8090 kHz. It allows continuous currents of up to 15 A_{rms} at generator voltages of less than 40 V_{rms}.
Application: Immunity test inductive charging 85 kHz
Datasheet NFCN 9732 85kHz


 Details
HHS 520616  Circular Helmholtz Coils


Number of turns: 
16 

Maximum Coil Current: 
55 A (5 min.) 

Nominal Coil Current: 
33 A continuous 

Coil Spacings (centered): 
300 mm 

Max. Magn. Field Strength: 
2100 A/m (5 min.) 

Nominal Magn. Field Strength: 
1260 A/m continuous 

Magn. Field Strength, 1 A Coil Current: 
38.17 A/m 151.63 dBµA/m


Coil Diameter: 
600 mm 

Max. Cubical shaped DuT: 
32.5 x 32.5 x 32.5 cm 

Mechanical Dimensions: 
0.64 m x 0.79 m x 0.42 m 

Terminals: 
4 mm female 

Inductance (Single Coil): 
340 µH 

Inductance (Coil Pair): 
750 µH 

Usable Frequency Range: 
DC  500 kHz 

Resonance Frequency: 
> 700 kHz 

Weight:

20.4 kg

Datasheet HHS 520616
Security Notes HHS 520616


 Details
HHS 5206132  Circular Helmholtz Coils


Number of turns: 
132 

Maximum Coil Current: 
15 A (5 min.) 

Nominal Coil Current: 
10 A continuous 

Coil Spacings (centered): 
300 mm 

Max. Magn. Field Strength: 
4713 A/m (5 min.) 

Nominal Magn. Field Strength: 
3142 A/m continuous 

Magn. Field Strength, 1 A Coil Current: 
314.2 A/m 169.97 dBµA/m


Coil Diameter: 
600 mm 

Max. Cubical shaped DuT: 
32.5 x 32.5 x 32.5 cm 

Mechanical Dimensions: 
0.64 m x 0.79 m x 0.42 m 

Terminals: 
4 mm banana jacks screw terminals


Inductance (Single Coil): 
22.4 mH 

Inductance (Coil Pair): 
49.9 mH 

Usable Frequency Range: 
DC  30 kHz 

Resonance Frequency: 
> 50 kHz 

Weight:

27.5 kg

Datasheet HHS 5206132


 Details
HHS 521010  Helmholtz Coils


Number of turns: 
10 

Maximum Coil Current: 
20 A (5 min.) 

Nominal Coil Current: 
10 A continuous 

Max. Magn. Field Strength: 
300 A/m (5 min.) 

Nominal Magn. Field Strength: 
150 A/m continuous 

Magn. Field Strength, 1 A Coil Current: 
15 A/m (Coil Dist. 0.4 m)


Mechanical Dimensions: 
1.0 x 1.0 x 0.63 m 

Coil Separation acc. IEC 100048: 
0.40 m 

Coil Separation for best uniformity: 
0.57 m 

Terminals: 
4 mm female 

Inductance (per Coil): 
0.36 mH 

Usable Frequency Range: 
DC  150 kHz 

Resonance Frequency: 
> 600 kHz 

Weight:

10 kg

Datasheet HHS 5210


 Details
HHS 5210100  Helmholtz Coils


Number of turns: 
100 

Maximum Coil Current: 
15 A (5 min.) 

Nominal Coil Current: 
9 A continuous 

Inner side length of the coil: 
1.0 m 

Electrical side length of the coil: 
1.04 m 

Maximum Coil Separation: 
0.70 m 

Usable Frequency Range: 
DC  10 kHz 

Inductance (per Coil): 
31.3 mH 

Resonance Frequency (Pair of Coils): 
>25 kHz 

Weight: 
36.5 kg 

Terminals: 
4 mm female 







The following data is valid for 0.4 m coil distance: 




Coil Separation acc. IEC 100048: 
0.40 m 

Max. Magn. Field Strength: 
2183 A/m (5 min.) 

Nominal Magn. Field Strength: 
1310 A/m continuous 

Magn. Field Strength, 1 A Coil Current: 
145.54 A/m (Coil Dist. 0.4 m)








The following data is valid for 0.566 m coil distance: 




Coil Separation for best uniformity along an axes through the coil center: 
0.566 m 

Magn. Field Strength, 1 A Coil Current: 
124.66 A/m (Coil Dist. 0.566 m) 

Recommended Accessories: 
NFCN 9732120 
Datasheet HHS 5210100
Recommended Accessories:


NFCN 9732120  Compensation Network


The compensation network NFCN 9732120 consists of a capacitor and a fuse in series. It is used for a series compensation of the HHS 5210100 at energetical mains frequencies of 50 Hz to 60 Hz.
The compensation capacitor reduces the total impedance of the series circuitry consisting of the Helmholtz coils HHS 5210100 and the NFCN 9732120 at the operating frequencies of 50 to 60 Hz. It allows continuous currents of up to 8 A rms at generator voltages of less than 70 V_{rms} ...80 V_{rms} .
Datasheet NFCN 9732120


 Details
HHS 5210100 2.5  Helmholtz Coils


Number of turns: 
100 

Maximum Coil Current: 
20 A (5 min.) 

Nominal Coil Current: 
15 A continuous 

Inner side length of the coil: 
1.0 m 

Electrical side length of the coil: 
1.04 m 

Maximum Coil Separation: 
0.70 m 

Usable Frequency Range: 
DC  10 kHz 

Inductance (per Coil): 
31.3 mH 

Resonance Frequency (Pair of Coils): 
>25 kHz 

Weight: 
56 kg 

Terminals: 
4 mm female 







The following data is valid for 0.4 m coil distance: 




Coil Separation acc. IEC 100048: 
0.40 m 

Max. Magn. Field Strength: 
2900 A/m (5 min.) 

Nominal Magn. Field Strength: 
1746 A/m continuous 

Magn. Field Strength, 1 A Coil Current: 
145.54 A/m (Coil Dist. 0.4 m)








The following data is valid for 0.566 m coil distance: 




Coil Separation for best uniformity along an axes through the coil center: 
0.566 m 

Magn. Field Strength, 1 A Coil Current: 
124.66 A/m (Coil Dist. 0.566 m) 
Datasheet HHS 5210100 2.5


 Details
HHS 521210  Helmholtz Coils


Number of turns: 
10 

Maximum Coil Current: 
20 A (5 min.) 

Nominal Coil Current: 
10 A continuous 

Maximum Coil Separation: 
0.70 m 

Usable Frequency Range: 
DC  150 kHz 

Inductance (per Coil): 
0.45 mH 

Resonance Frequency (Pair of Coils): 
>500 kHz 

Weight: 
12 kg 

Terminals: 
4 mm female 

Coil Separation acc. IEC 100048: 
0.48 m 

Coil Separation for best uniformity: 
0.68 m 

Max. Magn. Field Strength: 
250 A/m (5 min.) 

Nominal Magn. Field Strength: 
125 A/m continuous 

Magn. Field Strength, 1 A Coil Current: 
12.45 A/m (Coil Dist. 0.48 m)

Datasheet HHS 5212


 Details
HHS 521350  Helmholtz Coils


Number of turns: 
50 

Maximum Coil Current: 
8 A (5 min.) 

Nominal Coil Current: 
5 A continuous 

Max. Magn. Field Strength: 
390 A/m (5 min.) 

Nominal Magn. Field Strength: 
240 A/m continuos 

Magn. Field Strength, 1 A Coil Current: 
48.5 A/m (coil Dist. 0.75 m) 

Coil Separation acc. IEC 100048: 
0.75 m 

Maximum Coil Separation: 
0.82 m 

Mechanical Dimensions: 
1.36 m x 1.36 m x 0.9 m 

Usable Frequency Range: 
DC  20 kHz 

Inductance (per Coil): 
10.5 mH 

Resonance Frequency (Pair of Coils): 
>30 kHz 

Weight: 
18 kg 

Terminals: 
4 mm female 
Datasheet HHS 521350


 Details
HHS 5213100  Helmholtz Coils


Number of turns: 
100 

Maximum Coil Current: 
15 A (5 min.) 

Nominal Coil Current: 
9 A continuous 

Nominal Magn. Field Strength: 
900.9 A/m continuos 

Magn. Field Strength, 1 A Coil Current: 
100.096 A/m (coil Dist. 0.705 m) 

Maximum Coil Separation: 
0.83 m 

Mechanical Dimensions: 
1.46 m x 1.36 m x 0.91 m 

Usable Frequency Range: 
DC  5 kHz 

Inductance (Coil pair): 
88.3 mH (Coil Dist. 0.705 m) 

Resonance Frequency (Pair of Coils): 
>13 kHz 

Weight: 
44.5 kg 

Terminals: 
4 mm female 
Datasheet HHS 5213100


 Details
HHS 521510  Helmholtz Coils


Number of turns: 
10 

Maximum Coil Current: 
20 A (5 min.) 

Nominal Coil Current: 
10 A continuous 

Max. Magn. Field Strength: 
200 A/m (5 min.) 

Nominal Magn. Field Strength: 
100 A/m continuos 

Magn. Field Strength, 1 A Coil Current: 
9.96 A/m (coil Dist. 0.6 m) 

Coil Separation acc. IEC 100048: 
0.6 m 

Maximum Coil Separation: 
0.86 m 

Mechanical Dimensions: 
1.5 m x 1.5 m x 0.9 m 

Usable Frequency Range: 
DC  100 kHz 

Inductance (Coil pair): 
1.3 mH 

Resonance Frequency (Pair of Coils): 
>400 kHz 

Weight: 
14 kg 

Terminals: 
4 mm female 
Datasheet HHS 5215


 Details
HHS 5215100  Helmholtz Coils


Number of turns: 
100 

Maximum Coil Current: 
20 A (5 min.) 

Nominal Coil Current: 
10 A continuous 

Max. Magn. Field Strength: 
2000 A/m (max. 5 min., dist. 0.84 m*) 

Nominal Magn. Field Strength: 
842 A/m continuos 

Magn. Field Strength, 1 A Coil Current: 
84.2 A/m (coil dist. 0.84 m*) 

Maximum Coil Separation: 
0.93 m* 

Mechanical Dimensions: 
1.69 m x 1.60 m x 1.00 m 

Usable Frequency Range: 
DC  6 kHz 

Inductance (Coil pair): 
112 mH 

Resonance Frequency (Pair of Coils): 
>20 kHz 

Weight: 
53 kg 

Terminals: 
4 mm female 
Datasheet HHS 5215100


 Details
HHS 5218  Helmholtz Coils


Number of turns: 
10 

Maximum Coil Current: 
20 A (5 min.) 

Nominal Coil Current: 
10 A continuous 

Max. Magn. Field Strength: 
126 A/m (5 min.) 

Nominal Magn. Field Strength: 
63 A/m continuos 

Magn. Field Strength, 1 A Coil Current: 
6.26 A/m (coil Dist. 1.2 m) 

Coil Separation acc. IEC 100048: 
0.72 m 

Maximum Coil Separation: 
1.25 m 

Coil Separation for best uniformity: 
1.20 m 

Mechanical Dimensions: 
1.8 m x 1.8 m x 1.2 m 

Usable Frequency Range: 
DC  100 kHz 

Resonance Frequency (Pair of Coils): 
>400 kHz 

Weight: 
17 kg 

Terminals: 
4 mm female 
Datasheet HHS 5218


 Details
HHS 5230100  Helmholtz Coils

The HelmholtzCoil HHS 5230100 are especially designed to generate precisely defined magnetic fields from DC to ca. 5 kHz, e.g. according to SAE J55117.
Typical applications are magnetic immunity testing to whole vehicles according to SAEJ55117 (Vehicle Electromagnetic Immunity – Power Line Magnetic Fields).

Number of turns: 
100 

Maximum Coil Current: 
16 A (5 min.) 

Nominal Coil Current: 
8 A continuous 

Max. Magn. Field Strength: 
650 A/m @ coil Dist. =1.8 m 

Nominal Magn. Field Strength: 
325 A/m @ coil Dist. = 1.8 m 

Magn. Field Strength, 1 A Coil Current: 
50.8 A/m (coil Dist. 1.13 m) 

Maximum Coil Separation: 
>1.13 m 

Coil Separation for best uniformity: 
1.8 m 

Mechanical Dimensions: 
W:2.0 m x H:3.10 m x B:3.18 m 

Ground Clearence: 
0.12 m 

Usable Frequency Range: 
DC  5 kHz 

Resonance Frequency (per Coil): 
>10 kHz, typ. 13 kHz 

Weight (per Coil): 
190 kg 

Recommended Accessories: 
NFCN 9731100 
Datasheet HHS 5230100
Assembly HHS 5230100
Recommended Accessories:


NFCN 9731100  Compensation Network


The primary function of the compensation network NFCN 9731100 is to provide a serial compensation of the inductance of i.e. the HHS 5230100 at operation frequencies of 16.666 Hz, 50 Hz, 60 Hz, 150 Hz und 180 Hz. These operation frequencies are stipulated by SAE J551  17 (Vehicle Electromagnetic Immunity – Power Line Magnetic Fields).
The compensation capacitor reduces th e overall impedance of the series circuit consisting of the Helmholtz coil HHS 5230100 and the compensation network NFCN 9731100 at the applied operating frequency and allows currents up to 8 A_{rms} at a generator voltage of less than 60 V_{rms}.
Datasheet NFCN 9731100


 Details
HHS 3D 521350  3D HelmholtzCoil

With the threedimensional Helmholtz coil HHS 3D 521350 it is possible to create any vector field.

Number of turns (per Coil) 
50 

Maximum Coil Current: 
8 A (5 min.)


Nominal Coil Current: 
5 A continuos 

Maximum Magnetic Field Strength: 
390 A/m 

Nominal Magnetic Field Strength: 
240 A/m continuos 

Dimensions: 
1.4 m x 1.4 m x 1.3 m 

Frequency Range: 
DC  20 kHz 

Resonant Frequency (per Coil): 
> 30 kHz 

Weight: 
ca. 85 kg 

Inner Coil (YAxis)


Effective Coil Dimensions: 
1.1 m x 1.1 m 

Coil Separation acc. EN 55103: 
62 cm 

Magnetic Field Strength, 1 A: 
57.7 A/m 

Resistance (Coil pair): 
3.0 Ω 

Inductance (Coil pair): 
19.08 mH (Coil Dist. 0.62 m) 

Center Coil (XAxis) 

Effective Coil Dimensions:

1.21 m x 1.21 m 

Coil Separation acc. EN 55103: 
68 cm 

Magnetic Field Strength, 1 A: 
52.5 A/m 

Resistance (Coil pair): 
3.3 Ω 

Inductance (Coil pair): 
21.3 mH (Coil Dist. 0.68 m) 

Outer Coil (ZAxis) 

Effective Coil Dimensions: 
1.32 m x 1.32 m 

Coil Separation acc. EN 55103: 
74 cm 

Magnetic Field Strength, 1 A: 
48.2 A/m 

Resistance (Coil pair): 
3.6 Ω 

Inductance (Coil pair): 
23.58 mH (Coil Dist. 0.74 m) 
Datasheet HHS 3D 521350

